|Year : 2018 | Volume
| Issue : 4 | Page : 394-398
Comparison of Bone Marrow Biopsy Specimens Obtained Using a Motorized Device and Manual Biopsy Systems
Catherine A Glennon1, Janet M Woodroof2, Suman Kambhampati3, Alexis C Battershell4, Serena R O'Connor4, Kiley B Roberts4
1 The University of Kansas Cancer Center, Westwood, Kansas, USA
2 Department of Pathology, The University of Kansas Hospital, Kansas City, Kansas, USA
3 Sarah Cannon Cancer Institute, Research Medical Center, Kansas City, Missouri, USA
4 The University of Kansas Medical Center, Kansas City, Kansas, USA
|Date of Submission||06-Apr-2018|
|Date of Acceptance||04-Jun-2018|
|Date of Web Publication||10-Aug-2018|
Catherine A Glennon
Director, Patient and Community Education, The University of Kansas Cancer Center, Westwood, Kansas
Source of Support: None, Conflict of Interest: None
Objective: Bone marrow biopsy is an essential component in the diagnosis of hematopoietic disorders. Researchers evaluated the quality of bone marrow biopsy tissue acquired with a motorized bone marrow biopsy device versus a standard manual device based on the following criteria: biopsy length, percentage of aspiration artifact/intrastromal hemorrhage, length of nonhematopoietic bone, and overall quality of the sample. Methods: Bone marrow biopsies (motorized, n = 30; manual, n = 120) from two academic medical centers were evaluated by two board-certified hematopathologists. For each specimen, the following parameters were recorded: biopsy length (cm), aspiration artifact (assessed in intervals of ≤10%, 11%–25%, 26%–50%, 51%–75%, and >75%), length (cm) of nonhematopoietic biopsy (e.g., cortical bone and skin), and overall quality of sample (inadequate, suboptimal, adequate, and excellent). Results: Operators from two centers included physicians and nurse practitioners. The manual system was superior to the powered drill with respect to the amount of crush artifact (0.15 cm ± 0.01 vs. 0.24 cm ± 0.04, P = 0.01 [t-test]). There was a trend toward less aspiration artifact/intrastromal hemorrhage with the use of the manual biopsy; however, the difference was not statistically significant (P = 0.06). There was no statistically significant difference in the overall biopsy size, biopsy length, amount of nonhematopoietic elements, and overall adequacy of the sample. Conclusions: There was no significant difference in the biopsy length, amount of nonhematopoietic elements, and overall adequacy of the sample. Results suggest that the manual bone marrow biopsy device has significantly less crush artifact of the specimen and has a trend toward less aspiration artifact/intrastromal hemorrhage as well.
Keywords: Bone marrow aspiration and biopsy, crush artifact, manual approach, motorized device, pathology, quality
|How to cite this article:|
Glennon CA, Woodroof JM, Kambhampati S, Battershell AC, O'Connor SR, Roberts KB. Comparison of Bone Marrow Biopsy Specimens Obtained Using a Motorized Device and Manual Biopsy Systems. Asia Pac J Oncol Nurs 2018;5:394-8
|How to cite this URL:|
Glennon CA, Woodroof JM, Kambhampati S, Battershell AC, O'Connor SR, Roberts KB. Comparison of Bone Marrow Biopsy Specimens Obtained Using a Motorized Device and Manual Biopsy Systems. Asia Pac J Oncol Nurs [serial online] 2018 [cited 2021 Oct 24];5:394-8. Available from: https://www.apjon.org/text.asp?2018/5/4/394/236684
| Introduction|| |
Bone marrow biopsy is an integral part of the diagnosis, staging, and treatment management in patients with hematopoietic and nonhematopoietic disorders. It has been established that the quality of the biopsy has a role in the ability of pathologists to histologically identify and correctly diagnose a disease. Biopsy provides overall cellularity and information about detection of focal lesions, marrow cellularity, tumor metastasis, and the detection of disease.,
In particular, the length of the biopsy (a surrogate marker of quality) has been recommended to be at least 20 mm in the assessment of staging and response in non-Hodgkin's lymphoma patients. However, the biopsy length is often suboptimal due to various factors such as operator experience and patients' tolerance of the procedure. In hopes of improving the quality of bone marrow specimens obtained, researchers evaluated the use of the motorized biopsy device.
The motorized biopsy device is a battery-powered drill used to perform bone marrow aspiration and biopsy procedures. The device is designed to improve the overall experience by decreasing procedure time, reducing pain associated with the procedure, and decreasing physical effort required by clinicians as compared to a manual approach. A previous pilot study evaluating the motorized device resulted in overall positive feedback from the participants in all areas assessed, namely patient satisfaction, decreased pain, provider satisfaction, quality of specimen, and number of complications. Additional studies also report decrease in pain with the use of a motorized device.,
The greater the length, width, and volume of the core biopsy specimen, the greater the accuracy of diagnosis will be made. Overall, quality of aspirate specimen can be affected by hemodilution and the presence of crush artifact and hemorrhage in the sample. The evaluation of marrow is imperative in discerning the efficacy of treatment and providing information on the current disease status. If a diagnosis cannot be made due to the poor quality of a specimen, the procedure will most likely need to be repeated until an accurate diagnosis can be obtained.
Several prospective, randomized controlled trials have previously been reported showing that biopsy lengths obtained by the drill have been longer.,,, Few other studies have been completed with thorough end points, other than length, detailing results on specimen quality. One study revealed similar biopsy quality between manual and motorized devices with the presence of crush artifact, while another did not indicate a presence of crush artifact., Similarly, in the pilot study, an increased amount of focal intrastromal hemorrhage was identified, prompting a need for further investigation.
Although many studies show an increased specimen size with the use of a motorized device, there is little evidence showing the overall quality of the sample when comparing a motorized device with the manual method. The goal of this study was to retrospectively evaluate the bone marrow biopsies and assess the quality of the samples based on biopsy length, amount of nonhematopoietic elements, crush artifact, and aspiration artifact/intrastromal hemorrhage.
| Methods|| |
Research setting and population
The study received ethical clearance and permission from the university and hospital's Institutional Review Board and was in compliance with all ethical standards set by the committee. Each individual participated voluntarily; informed consent was obtained. Bone marrow biopsies (motorized, n = 30; manual, n = 120) were obtained and analyzed at an academic medical center. Inclusion criteria for both groups were patients who needed a bone marrow aspiration and biopsy as noted in the physician order. Both groups excluded patients with multiple myeloma.
The motorized device used was a battery-powered biopsy drill (11G × 103 mm). The manual device used was a 2-needle technique involving the T-Lok™ Bone Marrow Biopsy Needle (11G × 4”) and the Argon Bone Marrow Aspiration Needle (15G × 4”). For both approaches, lidocaine (Xylocaine-MPF) was injected locally, as conscious sedation is not utilized in this setting. Each participant started with 50 mg/5 mL of 1% solution. The average amount of lidocaine used was 5–10 ccs with the maximum amount of 30 ccs used.
For the manual approach, physicians and nurse practitioners performed the 2-needle procedure. The practitioner applied Xylocaine-MPF and used a small surgical blade to make a skin incision to insert the bone marrow aspiration needle. Once the needle contacted the bone, it was advanced by slow manual rotation until penetrating the cortical bone and entering the marrow cavity. A syringe collected a small amount of bone marrow aspiration.
For the motorized approach, a trained nurse practitioner followed the manufacturer directions for use of the motorized device. The same amount of Xylocaine-MPF was used with a manual procedure. The motorized device utilizes a cannula design (11G needle) to capture and hold bone marrow specimen from the soft-bone tissue. When using the motorized device, the procedure is completed by withdrawing marrow from the same anatomical location, the iliac crest. The procedure was performed in accordance with device directions for use, resulting in retrieving bone marrow aspiration to be analyzed by pathology.
Data collection and analysis
Two nurse practitioners obtained all of the motorized device specimens, while the manual specimens had been obtained previously by physicians, fellows, and nurse practitioners. Two board-certified hematopathologists evaluated the biopsies. For each specimen, researchers developed a two-page comprehensive evaluation tool [Table 1] and the following end points were assessed and recorded: biopsy length (cm), aspiration artifact (assessed in intervals of ≤10%, 11%–25%, 26%–50%, 51%–75%, and >75%), length (cm) of nonhematopoietic biopsy (e.g., cortical bone and skin), and overall quality of sample (inadequate, suboptimal, adequate, and excellent).
Two board-certified hematopathologists came to a consensus on judging the end points using a multiheaded microscope. All specimens retrieved by the motorized device were evaluated immediately upon biopsy. The specimens retrieved by manual technique were retrospectively selected and evaluated to serve as the comparison group.
| Results|| |
Researchers entered the data into Excel and manually calculated the demographic results and P values. The determined descriptive statistics for each demographic variable are summarized in [Table 2]. P < 0.05 was considered statistically significant.
The power drill showed a slight increase, yet no statistical significance, in the size of biopsy specimen over the manual procedure [(1.23 ± 0.09) cm vs. (1.09 ± 0.04) cm, P = 0.33]. Overall, there were less nonhematopoietic elements present in the power drill specimens compared to the manual procedure, but there was no statistical significance [(0.15 ± 0.03) cm vs. (0.20 ± 0.02) cm]. [Table 3] summarizes the diagnostic adequacy ratings.
Data indicate a statistically significant increase in the average amount of crush artifact present in the power drill specimens versus manual [(0.22 ± 0.04) cm vs. (0.15 ± 0.01) cm, t-test of P = 0.01]. This study revealed no statistically significant difference between the manual biopsy procedure and powered drill, in respect to biopsy size, aspiration artifact, intrastromal hemorrhage, amount of nonhematopoietic elements, or overall quality of specimen. However, the study results indicate a statistically significant difference in crush artifact, with a decreased amount in the manual procedure, thus indicating the manual method to be more superior when measuring crush artifact.
| Discussion|| |
There was no significant difference in the biopsy length, amount of nonhematopoietic elements, and overall adequacy of the sample. Results of this trial suggest that the manual bone marrow biopsy device has significantly less crush artifact and shows a trend toward less aspiration artifact/intrastromal hemorrhage as well. Given the results, the motorized bone marrow biopsy device produced samples that were not significantly different from samples retrieved via the manual method.
These findings were not consistent with past research with regard to significantly larger sample sizes.,,, In addition, some past studies found the manual technique to produce on an average more evaluable marrow specimens; however, this study found the two techniques to be very comparable. Comparable samples are in line with one other researcher's findings, which also identified crush artifact in the motorized device specimens. This specific motor device's high-powdered speed paired with the internal thread, located in the needle, has potential to damage and decrease the overall specimen size. These high-powered speeds create ease for the operator but present a potential danger for neighboring tissues if the needle were to slip off the bone.
Examination of marrow is essential for determining the stage of cancer and appropriate treatment for those who are newly diagnosed, as well as those undergoing treatment, with hematopoietic and nonhematopoietic diseases. Procedures and devices that yield quality specimens are of the utmost importance. Advanced practice nurses may perform bone marrow procedures; therefore, the use of a motorized device is a viable option. Previous studies indicate provider satisfaction with the motorized device and the time required to perform the procedure with decreased pain level reports from patients.,, When evaluating tools and techniques to use for bone marrow aspiration and biopsy, the use of a motorized device can be considered.
The small sample size within the motorized device group can be viewed as a limitation due to an inability to generalize the results. In addition, the number of practitioners completing the procedures presents another limitation. Even though all practitioners received the same training and instructions on how to complete the procedures, it cannot be assumed that there is no variation among technique, pressure, speed, etc., thus an interoperator variation can be considered as a limitation. Strengths include the large sample size within the group receiving the manual approach. In addition, the variety of end points assessed for each specimen make this study robust in the investigation of the quality of specimens retrieved.
Further research is needed to determine if the use of a motorized device increases the quality of bone marrow aspiration and biopsy specimens. Research involving a greater number of patients would be beneficial to analyze the quality of each specimen compared to the manual approach and ensure statistical relevance. It is also important to look at patient and clinician preferences, cost efficacy, time efficiency, and, most importantly, the accuracy of results when determining which approach is best. For clinicians, such as advanced practice nurses, who do multiple bone marrow procedures per day, use of the drill to decrease risks of potential injury from repetitive motion associated with the manual device, should be reviewed. Many previous studies have looked at pain and anxiety related to bone marrow aspirations.,,,, Further research can be completed to compare pain and anxiety levels using the motorized bone marrow biopsy device versus manual approach. In order to determine the superior procedural approach, all variables contributing to patient and clinician experience should be considered for future research implications.
The manuscript has been reviewed and approved by all authors reflecting honest work and the requirements for authorship have been met. Gratitude extended to Mary Luder, APN; Alison Holmes, RN; and other procedure staff involved with patient care and obtaining specimens.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Riley RS, Hogan TF, Pavot DR, Forysthe R, Massey D, Smith E, et al.
A pathologist's perspective on bone marrow aspiration and biopsy: I. Performing a bone marrow examination. J Clin Lab Anal 2004;18:70-90.
Campbell JK, Matthews JP, Seymour JF, Wolf MM, Juneja SK; Australasian Leukaemia Lymphoma Group, et al.
Optimum trephine length in the assessment of bone marrow involvement in patients with diffuse large cell lymphoma. Ann Oncol 2003;14:273-6.
Reed LJ, Raghupathy R, Strakhan M, Philbeck TE, Kim MY, Battini R, et al.
The OnControl bone marrow biopsy technique is superior to the standard manual technique for hematologists-in-training: A prospective, randomized comparison. Hematol Rep 2011;3:e21.
Gilotra M, Gupta M, Singh S, Sen R. Comparison of bone marrow aspiration cytology with bone marrow trephine biopsy histopathology: An observational study. J Lab Physicians 2017;9:182-9.
] [Full text]
Cheson BD, Horning SJ, Coiffier B, Shipp MA, Fisher RI, Connors JM, et al.
Report of an international workshop to standardize response criteria for non-Hodgkin's lymphomas. NCI sponsored international working group. J Clin Oncol 1999;17:1244.
Miller LJ, Philbeck TE, Montez DF, Puga TA, Brodie KE, Cohen SC, et al.
Powered bone marrow biopsy procedures produce larger core specimens, with less pain, in less time than with standard manual devices. Hematol Rep 2011;3:e8.
Glennon C. Evaluation of OnControl Powered bone Access System by Vidacare. The University of Kansas Cancer Center Annual Report; 2013. p. 7-9.
Berenson JR, Yellin O, Blumenstein B, Bojanower D, Croopnick J, Aboulafia D, et al.
Using a powered bone marrow biopsy system results in shorter procedures, causes less residual pain to adult patients, and yields larger specimens. Diagn Pathol 2011;6:23.
Voigt J, Mosier M. A powered bone marrow biopsy system versus manual methods: A systematic review and meta-analysis of randomised trials. J Clin Pathol 2013;66:792-6.
Swords RT, Anguita J, Higgins RA, Yunes AC, Naski M, Padmanabhan S, et al.
A prospective randomised study of a rotary powered device (OnControl) for bone marrow aspiration and biopsy. J Clin Pathol 2011;64:809-13.
Bucher CM, Lehmann T, Tichelli A, Tzankov A, Dirnhofer S, Passweg J, et al.
Comparison of a powered bone marrow biopsy device with a manual system: Results of a prospective randomised controlled trial. J Clin Pathol 2013;66:24-8.
Lee RK, Ng AW, Griffith JF. CT-guided bone biopsy with a battery-powered drill system: Preliminary results. AJR Am J Roentgenol 2013;201:1093-5.
Cohen SC, Gore JM. Evaluation of a powered intraosseous device for bone marrow sampling. Anticancer Res 2008;28:3843-6.
Lynch DW, Stauffer SL, Rosenthal NS. Adequacy of powered vs. manual bone marrow biopsy specimens: A retrospective review of sequential marrow aspirates and biopsies in 68 patients. Am J Clin Pathol 2015;143:535-9.
Islam A. Bone marrow solid core biopsy needle: A critical assessment of the utility, benefits and limitations of the instruments employed in current day haematology and oncology. J Clin Pathol 2018;71:475-82.
Yuvalı Karacan Y, Demircioğlu B, Ali R. Pain, anxiety, and depression during bone marrow aspiration and biopsy. Agri 2017;29:167-72.
Hjortholm N, Jaddini E, Hałaburda K, Snarski E. Strategies of pain reduction during the bone marrow biopsy. Ann Hematol 2013;92:145-9.
Brunetti GA, Tendas A, Meloni E, Mancini D, Maggiore P, Scaramucci L, et al.
Pain and anxiety associated with bone marrow aspiration and biopsy: A prospective study on 152 Italian patients with hematological malignancies. Ann Hematol 2011;90:1233-5.
Degen C, Christen S, Rovo A, Gratwohl A. Bone marrow examination: A prospective survey on factors associated with pain. Ann Hematol 2010;89:619-24.
Grønkjær M, Hasselgren CF, Østergaard AS, Johansen P, Korup J, Bøgsted M, et al.
Bone marrow aspiration: A Randomized controlled trial assessing the quality of bone marrow specimens using slow and rapid aspiration techniques and evaluating pain intensity. Acta Haematol 2016;135:81-7.
| Authors|| |
Catherine A. Glennon
[Table 1], [Table 2], [Table 3]